# Flow Pressure Transducers for Fluid Pipelines

Flow Pressure Transducers are measurements of fluid pressure within a pipeline. When fluid flows through a pipe, pressure acts on the pipe wall. The medium in the pipeline can be liquid or gas.
Measure the pressure of the fluid, on the one hand, you can understand the pressure in the pipeline. On the other hand, pipeline flow can be monitored through the relationship between pipe diameter-pressure-flow.

## Featured Flow Pressure Transducers for Sale

Extended reading: Static Pressure vs Dynamic Pressur

## Measuring Flow with Pressure Sensors

Differential pressure sensors have been widely used to measure the flow of incompressible liquids. A very common method is to measure the pressure drop across the orifice in the pipe and calculate the flow.

An orifice plate is simply a plate that fits in a pipe (usually between flanges) with a hole of known size in its center. As the fluid flows through the orifice, a pressure drop develops across the orifice plate from the upstream side to the downstream side. This pressure drop is proportional to flow, and the sensor signal can be used to calculate flow in engineering units.

The upstream side of the orifice will have higher pressure and will be connected to the + port of the pressure sensor through a three-valve manifold. The downstream side of the orifice is similarly connected to the -port of the pressure sensor. The three-valve manifold is used to protect the pressure sensor from overpressure when the pressure sensor is installed into the work pipeline.

Calculations to determine flow from pressure drop are derived from fairly simple physical equations. However, there are many variables, each with its own engineering unit. These include: hole geometry, pipe size, fluid viscosity and fluid density. Given the number of terms and the conversion factors involved in each variable, the calculation can be quite involved.

Fortunately, there are many online calculators that allow you to simply enter variables in any engineering units that are convenient for you, and then calculate the flow rate for any given pressure drop across the orifice.

Extended reading: Temperature and pressure compensation of vortex steam flowmeter

## Pipe diameter/velocity/flow comparison table (inlet > outlet)

 Diameter (DN) Flow m3/h 0.4m/s 0.6m/s 0.8m/s 1.0m/s 1.2m/s 1.4m/s 1.6m/s 1.8m/s 2.0m/s 2.2m/s 2.4m/s 2.6m/s 2.8m/s 3.0m/s 20 0.5 0.7 0.9 1.1 1.4 1.6 1.8 2.0 2.3 2.5 2.7 2.9 3.2 3.4 25 0.7 1.1 1.4 1.8 2.1 2.5 2.8 3.2 3.5 3.9 4.2 4.6 4.9 5.3 32 1.2 1.7 2.3 2.9 3.5 4.1 4.6 5.2 5.8 6.4 6.9 7.5 8.1 8.7 40 1.8 2.7 3.6 4.5 5.4 6.3 7.2 8.1 9.0 10.0 10.9 11.8 12.7 13.6 50 2.8 4.2 5.7 7.1 8.5 9.9 11.3 12.7 14.1 15.6 17.0 18.4 19.8 21.2 65 4.8 7.2 9.6 11.9 14.3 16.7 19.1 21.5 23.9 26.3 28.7 31.1 33.4 35.8 80 7.2 10.9 14.5 18.1 21.7 25.3 29.0 32.6 36.2 39.8 43.4 47.0 50.7 54.3 100 11.3 17.0 22.6 28.3 33.9 39.6 45.2 50.9 56.5 62.2 67.9 73.5 79.2 84.8 125 17.7 26.5 35.3 44.2 53.0 61.9 70.7 79.5 88.4 97.2 106.0 114.9 123.7 132.5 150 25.4 38.2 50.9 63.6 76.3 89.1 101.8 114.5 127.2 140.0 152.7 165.4 178.1 190.9 200 45.2 67.9 90.5 113.1 135.7 158.3 181.0 203.6 226.2 248.8 271.4 294.1 316.7 339.3 250 70.7 106.0 141.4 176.7 212.1 247.4 282.7 318.1 353.4 388.8 424.1 459.5 494.8 530.1 300 101.8 152.7 203.6 254.5 305.4 356.3 407.1 458.0 508.9 559.8 610.7 661.6 712.5 763.4 350 138.5 207.8 277.1 346.4 415.6 484.9 554.2 623.4 692.7 762.0 831.3 900.5 969.8 1039.1 400 181.0 271.4 361.9 452.4 542.9 633.3 723.8 814.3 904.8 995.3 1085.7 1176.2 1266.7 1357.2 450 229.0 343.5 458.0 572.6 687.1 801.6 916.1 1030.6 1145.1 1259.6 1374.1 1488.6 1603.2 1717.7 500 282.7 424.1 565.5 706.9 848.2 989.6 1131.0 1272.3 1413.7 1555.1 1696.5 1837.8 1979.2 2120.6 600 407.1 610.7 814.3 1017.9 1221.4 1425.0 1628.6 1832.2 2035.7 2239.3 2442.9 2646.5 2850.0 3053.6

 Diameter(DN) Recommended flow rate m/s: 20 25 32 40 50 65 80 100 125 150 200 250 300 350 400 closed system 0.5-0.6 0.6-0.7 0.7-0.9 0.8-1 0.9-1.2 1.1-1.4 1.2-1.6 1.3-1.8 1.5-2.0 1.6-2.2 1.8-2.5 1.8-2.6 1.9-2.9 1.6-2.5 1.8-2.6 open system 0.4-0.5 0.5-0.6 0.6-0.8 0.7-0.9 0.8-1.0 0.9-1.2 1.1-1.4 1.2-1.6 1.4-1.8 1.5-2.0 1.6-2.3 1.7-2.4 1.7-2.4 1.6-2.1 1.8-2.3

## Applications of Flow Pressure Transducers

Fluid Pressure Transducers are mainly used in: construction machinery, hydraulic and pneumatic systems, petrochemicals, energy and water treatment, water conservancy and hydropower, compressors and other fields:

1. Applied to hydraulic system
2. Applied to safety control system
3. Applied to injection mold
4. Applied to monitoring mine pressure
5. used in compressors, air conditioning and refrigeration equipment

Extended reading: High Pressure Flow Meters Types

## How to install the pressure sensor when measuring the flow pipeline?

When detecting the gas and fluid pressure in the pipeline, the pressure guiding pipe is connected with a vertical tee before entering the pressure sensor. The sampling gas enters from the upper part of the tee. The lower part of the tee is connected to a water collecting irrigation, and the horizontal interface of the tee enters the pressure sensor .

The pressure sensor is installed horizontally. The sampling pressure pipe is connected to the upper part of the pressure sensor detection diaphragm. The lower part of the detection diaphragm has its own discharge bolt.

When there is liquid entering the pressure guiding pipe, a large part will be collected in the water collecting irrigation. A small amount of liquid entering the pressure sensor detection diaphragm can be discharged by the discharge bolt of the pressure sensor.

When measuring fluid pressure, the pressure of the main fluid is taken as the measurement object. Therefore, when the pressure sensor measures the gas pressure in the pipeline, in order to prevent the liquid in it from entering the sampling pressure pipe. The sampling point is 45° above the horizontal plane of the center of the pipeline.

When measuring the liquid pressure of the pipeline, in order to prevent the gas in it from entering the sampling pressure guiding tube, the sampling point is 45° below the horizontal plane of the center of the pipeline. Read more about: Shop 101: Key Factors In Selecting A Pipe Flow Meter

Even so, it is inevitable that different phases of fluid will enter the sampling impulse pipe. There are different requirements for the installation and configuration of the pressure sensor impulse pipe.

The water collecting tank or gas collecting tank installed on the pressure guiding pipe is equipped with an inlet valve and a discharge valve. During normal testing, the inlet valve is opened and the discharge valve is closed. When it is necessary to discharge, first close the inlet valve, and then open the discharge valve. . This discharge method will always have a bit of out-of-phase fluid remaining in the water or gas header, but it will not affect the pressure sensor detection accuracy.

Fluids include both gaseous fluids and liquid fluids. The density and pressure of gaseous fluids are lower than that of liquid fluids. So in pipeline fluids, the gaseous fluid is always above the liquid fluid. This is the basic characteristic of vapor-liquid two-phase fluid.

Extended reading: Pipe Diameter vs Pressure vs Flow

Pressure transmitters can be categorized into four main types:

• Gauge Pressure Transmitter.
• Absolute Transmitters.
• Differential-Pressure Transmitter.
• Multivariable Pressure Transmitters.

Extended Reading: 4 wire pressure sensor wiring diagram

There are many types of pressure sensors. Such as resistance strain gauge pressure sensors, semiconductor strain gauge pressure sensors, piezoresistive pressure sensors, inductive pressure sensors, capacitive pressure sensors, resonant pressure sensors, etc.

Extended reading: wireless pressure transmitter working principle

Currently, the most widely used pressure sensors include:

• Diffused silicon piezoresistive pressure sensor,
• Ceramic piezoresistive pressure sensor,
• Sputtering thin film pressure sensor,
• Capacitive pressure sensor,
• Sapphire pressure sensor with high temperature resistance.
• But the most widely used is the piezoresistive pressure sensor, which has a very low price, high accuracy and good linearity.

Extended Reading: Electronic Pressure Switch for Air Compressor

Flow is an important parameter in industrial production and life. The amount of fluid flowing through a section of a pipe per unit time is called instantaneous flow. Instantaneous flow is divided into volume flow and mass flow. A flow sensor is a sensor that measures the flow of a fluid. Flow sensor is an important instrument in measurement technology, it is widely used in industrial process control, life science and technology, commercial application, military and other fields. In recent years, with the development of science and technology and the development of human needs, flow sensors are also constantly developing.

## Related Flow-Pressure Measurement Solutions

### Collection of Industrial Insertion Flowmeter Types

What is an insertion flow meter? Insertion flowmeter is a type of flowmeter in which the measuring probe is inserted into the pipe to measure the flow. This is a…

### Crude Oil Flow Meter

Crude Oil Flow Meter refers to a type of flow meter that can accurately monitor and measure the flow of crude oil. Crude oil is an industrial raw material with…

### Pressure Transducer Wiring: 2 Wire-3 Wire-4 Wire

Pressure Transducer Wiring refers to the electrical connection of the pressure transmitter. A pressure transmitter is a conversion device that converts pressure signals into analog signals or digital signals. Generally,…

### High Accuracy Pressure Transducers

High Accuracy Pressure Transducers are also called high precision pressure transducers or high accuracy pressure sensors. Sino-Inst manufactures various types of High Accuracy Pressure Transducers. From the perspective of accuracy,…

### Flat Pressure Sensor for High Viscosity Fluids

Flat Pressure Sensor is a type of pressure sensor that specifically refers to the measurement diaphragm without pressure holes. Also known as a flat film pressure sensor. Flat Pressure Sensor…

### Non Contact Flow Meters Measure Liquid Flow

What is non contact flow meter? Non Contact Flow Meters refers to flow meters that can achieve flow measurement without contacting the fluid medium. No need to destroy the pipeline…

### What Is a Fluid Pressure Sensor?

What Is a Fluid Pressure Sensor? Fluid Pressure Sensors refers to sensors that can be used to measure the pressure of liquid media. Such as measuring pressure of Oil, Fuel…

### What Is Difference Between Rotameter and Flow Meter?

What is a rotameter flow meter? Rotometer, also known as Float Type Flowmeter, is widely used in industry. It can measure the flow of liquid, gas and steam, and it…

### Mechanical Oil Flow Meters

What is mechanical oil flow meter? Mechanical oil flow meter refers to a flowmeter that can detect the volume flow of oil inline. Most importantly, this type of flowmeter is…

### What Is a Smart Pressure Transmitter?

What Is a Smart Pressure Transmitter? Smart pressure transmitter is also called Smart pressure sensor or Smart pressure transducer. Smart pressure transmitter is the best pressure measuring instrument. It is…

### Top 8 Types Petroleum Flow Meters for Oil and Gas Industry

In the new era, petrochemical enterprises are facing unprecedented challenges. How to improve the production efficiency of the petroleum industry and improve the product quality is an important problem that…

### High Temperature Pressure Transducer with Best Price

High Temperature Pressure Transducer is also called High Temperature Pressure Sensor, or High Temperature Pressure Transmitter. The High Temperature Pressure Transducer is used for applications where the medium temperature exceeds…

### Clamp on Water Flow Meter

What is Clamp on Water Flow Meter? Clamp on Water Flow Meter is also known as the clamp-on ultrasonic flowmeter. Using an ultrasonic flowmeter to measure water flow rate is…

### Slurry Flow Meter

Flow Meter for Slurry Application Slurry is a product of domestic sewage and industrial wastewater treatment. Slurry consists of solid impurities, a slurry of suspended substances. The main characteristic of…

### Measuring Steam Flow and Steam Flow Meters

Measuring Steam Flow Measuring Steam Flow is a difficult task. But again, it’s a very important thing. Steam flow measurement is difficult, the main reason is determined by the characteristics…

### Fuel Flow Meter for Boat-Selection & Application

Fuel Flow Meter for Boat refers to a flow meter dedicated to measuring marine fuel. There are many types of marine fuel oil. Mainly include light diesel, heavy diesel, fuel…

### Guide for Digital Fuel Flow Meter

Digital Fuel Flow Meter is a flow meter dedicated to measuring fuel, diesel, gasoline, and petroleum. Digital Fuel Flow Meter generally has a digital display or signal output. Such as…

Extended reading: High accuracy pressure transducers

Sino-Inst is Manufacturer of Flow Pressure Transducers for Fluid Pipelines. For High Viscosity Fluids-Chemical Coatings – Paints – Slurry – Asphalt – Crude Oil. We supply more than 20 kinds of Flow Pressure Transducers.

Flow Pressure Transducers are mainly used for highly viscous media.
It can measure even liquids such as water, seawater, oil, and slurry.

Sino-Inst’s Flow Pressure Transducers, made in China, Having good Quality, With better price. Our pressure measurement instruments are widely used in China, India, Pakistan, the US, and other countries.

The entire team at Sino-Inst’s has received excellent training, so we can ensure that every client’s needs are met. For assistance with your product requirements, whether it’s a Flow Pressure Transducer, level sensor, or other device, give us a call.

Request a Quote