Sino-Inst is one of the most reliable Turbine flow meter manufacturers and suppliers in China.
A turbine flow meter measures volumetric fluid flow in a closed pipe. Turbine flow meters measure the flow rate of liquids, gases and vapors in pipes,
such as natural gas, chemicals, water, cryogenic liquids, air, and industrial gases. Digital turbine flow meters can work as the eletromagnetic flow meters, and mass flow meter, for the inline flow measurement. But the turbine flow meter accuracy is the highest.
Inline turbine flow meters
What is a turbine flow meter?
Turbine flow meters are devices,that measure fluid flow through the use of axial turbines.
There are a variety of different ways to categorize flow meters,
and fluid flow measurements.
The two main flow meter categories are volumetric and mass flow meters,
which separate flow meters based on whether they measure flow as a volume per unit time or a mass per unit time.
The categories can be broken down further,
depending on the construction of the meter and the method used to make the measurement.
These categories include pressure-based meters, mechanical flow meters,
orifice plate flow meters, and many others.
Turbine flow meters fall into the volumetric flow meter and mechanical flow meter categories.
The main component of the turbine flow meter is, naturally, the axial turbine.
The turbine is typically located at the center of the pipe with its axis aligned with the axis of the pipe.
As a liquid or gas passes through the pipeline,
it comes in contact with the turbine and imparts some of its energy on the blades of the turbine.
The force of the fluid on the turbine blades causes the turbine to spin,
and the speed at which the turbine spins can be used to determine the flow rate of the liquid or gas.
Faster flowing fluids exert greater forces on the turbine blades,
and thus cause the turbine to spin faster,
and slower moving fluids impart less energy on the turbine,
causing the turbine to spin slower.
This relationship between flow rate and turbine speed can be expressed using a formula,
which is used to accurately calculate the fluid flow rate.
Flow meter turbines can be manufactured from a variety of different materials,
to better fit the needs of specific applications.
Plastics and non-corrosive metal alloys are popular materials,
for use with corrosive and volatile fluids,
as they drastically improve the turbines longevity and performance.
Due to the mechanical nature of the meter,
an upstream filter is essential,
especially for use in applications involving inhomogeneous fluids.
The filter helps catch debris and impurities,
that could reduce the quality of the fluid product and helps prevent damage to the turbine.
Turbine flow meters are used across a broad spectrum of industries,
and are used in applications ranging from water distribution to fire protection.
Large commercial users and water utilities take advantage of these flow meters,
because they are affordable, easy to use,
and provide accurate flow rate measurements.
Additionally, the fire protection industry utilizes specialized turbine flow meters,
that are temporarily installed to meter the water used from fire hydrants.
While these flow meters are very accessible and effective, they do have limitations.
Compared to other non-mechanical flow meters,
maintenance is a much greater issue with turbine flow meters.
Filters can become clogged from an abundance of debris or impurities,
and turbines can be damaged from prolonged use.
Careful attention should be given to maintaining the meter to ensure,
that it continues to provide accurate measurements.
The accuracy of turbine flow meters is also heavily dependent on the fluid flow,
becoming inaccurate with lower flow rates.
But, with the proper precautions and understanding of your applications,
a turbine flow meter is an exceptional option for your flow metering needs.
Let’s see the turbine flow meter wikipedia.
Application of turbine flow meter
These sturdy, reliable, precise devices are used to measure the volumetric flow rate of gases and clean, low-viscosity liquids.
They are found throughout industry, but especially in the food and chemical sectors.
Turbine flowmeters are less accurate at low flow rates, due to rotor/bearing drag that slows the rotor.
Make sure to operate these flowmeters above approximately 5 percent of maximum flow.
Turbine flowmeters should not be operated at high velocity, because premature bearing wear and/or damage can occur.
Be careful when measuring fluids that are non-lubricating, because bearing wear can cause the flowmeter become inaccurate and fail.
In some applications, bearing replacement may need to be performed routinely, and increase maintenance costs.
Application in dirty fluids should generally be avoided, so as to reduce the possibility of flowmeter wear and bearing damage.
In summary, turbine flowmeters have moving parts, that are subject to degradation with time and use.
Extended reading: Turbine Insertion Flow Meter for Large Diameter Pipeline
Abrupt transitions from gas flow to liquid flow should be avoided, because they can mechanically stress the flowmeter, degrade accuracy, and/or damage the flowmeter.
These conditions generally occur when filling the pipe and under slug flow conditions.
Two-phase flow conditions can also cause turbine flowmeters to measure inaccurately.
You may like:
- Kerosene Gasoline/ Kerosene Flow Meter
- Water Level Transducers – Detailed Guide
- Upstream and Downstream Flow Straight Pipe Requirements-for Flowmeter Installation
- Explosion Proof Ultrasonic Level Sensor| Hazardous Area-Non Contact
- Slurry Flow Meter
- Measuring Steam Flow and Steam Flow Meters
- New Trend of Transformer Oil Level Indicator
- External Tank Level Indicator
- 80GHZ Radar Level Transmitter
- Material Level Indicators
- Digital Flow Meter for Argon Gas
Sino-Inst’s Turbine Meters,
offer reliable and accurate volume flow measurement,
in a large range of sizes and pressure rating,
while being fully compliant with international standards.
Sino-Instrument is the risk-free choice for your gas flow measurement applications.
Sino-Instrument, supplies turbine flow meters,
like: gas turbine flow meter, gpi turbine flow meter,
liquid turbine flow meter, sanitary turbine flow meter,
insertion turbine flow meter, steam turbine flow meter,
and natural gas turbine flow meter.
Sino-Inst ’s turbine flow meters, made in China. Having good Quality, With better price.
Our flow measurement instruments are widely used in China, India, Pakistan, US, and other countries.
Request a Quote