Ultrasonic level measurement-
Continuous level measurement in liquids and solids with ultrasonic level sensors
Ultrasonic Level Transmitters provide non-contact and maintenance-free level measurement, (Like the Radar )
for fluids, pastes, sludges and powdery to coarse bulk materials.
Ultrasonic level measurement, easy installation, wireless, portable and can be explosion-proof.
Sonic is the sound we can hear. Ultrasonic is the sound above the human hearing range.
A human can hear maximum up to a frequency of 20 KHz.
Ultrasonic frequencies are above 20 KHz.
Ultrasonic waves are used to measure the level of liquids and solid objects in industries.
Ultrasonic level measurement is the contactless principle and most suitable for level measurements of hot, corrosive and boiling liquids.
The normal frequency range used for ultrasonic level measurements is within a range of 40 200 KHz.
You may like: Magnetostrictive level sensor
1. What is the principle of ultrasonic level measurement?
Ultrasonic waves detect an object in the same way as Radar does it.
Ultrasonic uses sound waves, and Radar uses radio waves.
When the ultrasonic pulse signal is targeted towards an object,
it is reflected by the object and echo returns to the sender.
The time traveled by the ultrasonic pulse is calculated,
and the distance of the object is found.
Bats use a well-known method to measure the distance while traveling.
The ultrasonic level measurement principle is also used to find out fish positions in the ocean,
locate submarines below water level,
also the position of a scuba diver in the sea.
We will refer to Fig-1 and make an effort to understand the technicalities of the ultrasonic level transmitter.
An ultrasonic level transmitter is fixed at the top of a tank half-filled with liquid.
The reference level for all measurements is the bottom of the tank.
Level to be detected is marked as “C”, and “B” is the distance of the ultrasonic sensor from the liquid level.
Ultrasonic pulse signals are transmitted from the transmitter, and it is reflected back to the sensor.
Travel time of the ultrasonic pulse from the sensor to target and back is calculated.
Level “C” can be found by multiplying half of this time with the speed of sound in air.
The measuring unit final result can be centimeters, feet, inches, etc.
Level = Speed of sound in air x Time delay / 2
|Tip||You can see our range of Ultrasonic Level Transmitters here.|
2. Practical system design problems of Ultrasonic Level Transmitter:
The above principle of measurement looks quite straightforward and true only in theory.
In practice, there are some technical difficulties which are to be taken care to get correct level reading.
- The velocity of sound changes due to the variation of air temperature.
- An integrated temperature sensor is used to compensate for changes in the velocity of sound due to temperature variations.
- There are some interference echoes developed by the edges, welded joints, etc.
- This is taken care of by the software of the transmitter and called interference echo suppression.
- Calibration of the transmitter is crucial.
- Accuracy of measurement depends on the accuracy of the calibration.
- The Empty distance “A” and measurement span “D” is to be ascertained correctly for inclusion in the calibration of the transmitter.
- The transient characteristics of the sensor will develop a Blocking distance as shown in Fig-1. Span “D” should never extend to the blocking distance.
3. The basic structure of an Ultrasonic Transducer (Refer: Fig-2):
Ultrasonic Sensor is the heart of the ultrasonic level Transmitter instrument.
This sensor will translate electrical energy into ultrasound waves.
Piezoelectric crystals are used for this conversion process.
Piezoelectric crystals will oscillate at high frequencies when electric energy is applied to it.
The reverse is also true.
These piezoelectric crystals will generate electrical signals on receipt of ultrasound.
These sensors are capable of sending ultrasound to an object and receive the echo developed by the object.
The echo is converted into electrical energy for onward processing by the control circuit.
4. Functional block diagram of a typical Ultrasonic Level Transmitter:
We will refer to Fig-3 Functional Block Diagram to clarifying physical structures of an Ultrasonic Level Transmitter.
A micro-controller based Control Circuit monitors all the activities of the ultrasonic level transmitter.
There are two Pulse Transmission Circuits, one for transmitter pulse and the other one for receiver pulse.
The pulse generated by the transmitter pulse is converted to Ultrasound pulses,
by the Ultrasonic Sensor(Transmitter) and targeted towards the object.
This ultrasound pulse is reflected back as an echo pulse to the Ultrasonic Sensor (Receiver).
The receiver converts this Ultrasonic pulse to an electrical signal pulse through the pulse generator.
The time elapsed, or the reflection time is measured by the counter.
This elapsed time has relation to the level to be measured.
This elapsed time is converted to level by the Control Circuit.
There is a Timing Generator Circuit which is used to synchronize all functions,
in the ultrasonic level measurement system.
The level is finally converted to a 4-20mA signal.
4mA is 0% level, and 20mA is the 100% level (see Fig-1).
This 4-20mA output signal carrying the level data can be transmitted to long distance to Process Control Instruments.
5. Advantages of Ultrasonic Level Transmitter:
The ultrasonic level transmitter has no moving parts,
and it can measure the level without making physical contact with the object.
This typical characteristic of the transmitter is useful for measuring levels in tanks with corrosive,
boiling and hazardous chemicals.
The accuracy of the reading remains unaffected even after changes,
in the chemical composition or the dielectric constant of the materials in the process fluids.
6. Limitations of Ultrasonic Level Transmitter:
Ultrasonic level transmitters are the best level measuring devices,
where the received echo of the ultrasound is of acceptable quality.
It is not so convenient if the tank depth is high or the echo is absorbed or dispersed.
The object should not be sound-absorbing type.
It is also unsuitable for tanks with too much smoke or high-density moisture.
ULT-200 Ultrasonic Level Detector
Ultrasonic Level Detector, also called ultrasonic level transmitter, or water level indicating transmitter.
ULT-200 ultrasonic level detector, non-contact type level transmitter,
measuring liquid level or solid applications up to 50 ft. / 15 meters,
or flow rates in all types of open channel flow applications.
ULT-100A Ultrasonic Level Transducer
When advanced communication functions and calculations are required
Ultrasonic Level Transducer, also called Ultrasonic level sensor, or ultrasonic level transmitter.
Ultrasonic level transmitter is water level transmitter,
which can measure Fluid level depth, oil tank level, with 4-20mA.
ULT-100 Ultrasonic Level Indicator
For remote monitoring. Send SMS alerts.
Ultrasonic Level Indicator, also called Ultrasonic level sensor, or ultrasonic level transmitter.
Ultrasonic Level Transmitter provides reliable, accurate, non-contact measurement of liquid levels.
ULT-200A Ultrasonic fuel tank level sensor
Ultrasonic fuel tank level sensor, also called non-contact fuel meter,
is an electronic level transmitter,
measuring continuous liquid level, fluid level, or oil level.
Sino-Instrument offers 20 ultrasonic level detectors products.
A wide variety of ultrasonic level detector options are available to you, such as free samples, paid samples.
Sino-Instrument is ultrasonic level transmitter suppliers, located in China.
Ultrasonic level detector products are most popular in North America, Mid East, and Eastern Europe.
The United States, and India, which export 99%, 1%, and 1% of ultrasonic level transmitter respectively.
You can ensure product safety by selecting from a certified supplier,
with ISO9001, ISO14001 certification.