Nitrogen gas flow meter is used to measure the flow rate of N2 gas. Digital vortex flowmeter can be flange-Wafer-threaded installation. Threaded is suitable for small diameter pipes.
Nitrogen gas flow meter is easy to use, reasonably priced. Threaded connection is a widely used detachable fixed connection. It has the advantages of simple structure, reliable connection, and convenient assembly and disassembly. Nitrogen is required in many processes of the chemical and pharmaceutical industry. Predominantly as protective gas, to prevent the occurrence of explosive atmospheres. Sino-Inst vortex flowmeter can be used for natural gas, nitrogen, biogas and other gases.
Vortex Nitrogen Gas Flow Meter reference price: USD 400-1000/pc.
Sino-Inst offers a variety of Vortex flow meters for flow measurement. If you have any questions, please contact our sales engineers.
Features of Vortex Nitrogen Gas Flow Meter
- The nitrogen flow can be controlled more precisely to save costs.
- The nitrogen flow control can also prevent the nitrogen flow from being too large, resulting in insufficient heat supplementation.
- The amount of nitrogen output can be accurately known.
- The nitrogen flow meter has no movable parts, no wear, and dirt resistance. No mechanical maintenance is required, and the service life is long;
- The overall structure design of the instrument is reasonable, the dynamic measurement range is wide, and the pressure loss is small;
- The measurement is not affected by temperature and pressure, and is resistant to high temperatures & High pressure.
Specifications of Vortex Nitrogen Gas Flow Meter
|《JB/T9249-2015 Vortex flow meter industry standard》|
|Nominal pressure(MPa)||DN15-DN200 4.0(>4.0,Supply agreement)DN250-DN300 1.6(>1.6,Supply agreement)|
|Medium temperature(ºC)||Piezoelectric type :-40~150,-40~260,-40~330;|
Projected Capacitive:-40~400(upply agreement)
|Body material||1Cr18Ni9Ti(Other material order by agreement)|
|Piezoelectric type :0.2g Projected Capacitive:1.0-2.0g|
|Range of degrees||1:6~1:25|
|Power Supply||Sensor:DC+12V,DC+24V;Transmitter:DC+12V,DV+24V;Battery-powered:3.6V battery|
|Output signal||Pulse:high level:≥5V low level:≤1V;electricity current:4~20mA|
|Pressure loss coefficient||Comply with JB/T9249 standard Cd≤2.4|
|Explosion-proof coefficient||Intrinsic safety :Ex ia II CT2-T4 Explosion-proof type:Exd II CT2-T5|
|Level of protection||Ordinary type:IP65 Diving type:IP68|
|Environmental conditions||Tempersature:-20ºC~55ºC,relative humidity 5%~90%,atmospheric pressure:86~106kPa|
|Applicable medium||gas ,liquid,steam|
|Transmission distance||Three-wire pulse output:≤300m,Two-wire standard current output:(4~20mA)≤1500m|
|Counduit entry||Thread M20*1.5 or other|
You may like: Digital Flow Meters Guide| Liquid and Gas
What is the difference between Wafer type and screw type connection vortex flowmeter?
Three connection methods: Wafer, thread, flange, etc.:
The flange connection can be selected according to the process requirements, according to different joint surfaces, pressure levels, etc.
Wafer type can be used for various pipe diameters, but the pressure level is not too high.
The application of threaded connection is relatively small, such as small diameter, medium and low pressure, etc.
Read more about: Liquid Flow Meters Guide
Five matters to pay attention to when choosing threaded Vortex Nitrogen Gas Flow Meter
The principle of the vortex flowmeter is to install a stagnant part in the flowmeter pipe. When the fluid flows through the stagnant part. Due to the stagnant effect on the surface of the stagnant part, there will be two asymmetric rows downstream.
These vortices are separated on the side and rear of the stagnant member to form a so-called Karman vortex train. The rotation directions of the two vortices are opposite. Karman theoretically proved that when h/L=0.281, the vortex train is stable. h is the width between two vortex rows, and L is the distance between two adjacent vortices
Correct selection can ensure the better use of threaded connection vortex flowmeter.
What kind of vortex flowmeter should be selected according to the physical and chemical properties of the fluid medium to be measured?
- Precise function inspection
Accuracy grade and function Choose the instrument accuracy grade according to the measurement requirements and use occasions, so as to be economical and cost-effective.
For example, it is used for trade settlement, product handover and energy metering occasions. The accuracy level should be higher, such as 1.0, 0.5, or higher.
For process control occasions, different accuracy levels can be selected according to control requirements.
For some occasions that just check the process flow and do not need to do precise control and measurement, you can choose a slightly lower accuracy level, such as 1.5, 2.5, or even 4.0. In this case, you can choose a cheap plug-in vortex flowmeter.
2. Measurable medium
When measuring general media, the full-scale flow of the vortex flowmeter can be selected within the range of 0.5-12m/s of the measured medium flow rate, and the range is relatively wide.
The selected instrument specification (caliber) is not necessarily the same as that of the process pipeline. It should be determined whether the measurement flow range is within the flow rate range. That is, when the flow rate of the pipeline is low and cannot meet the requirements of the flow meter or when the measurement accuracy cannot be guaranteed at this flow rate. It is necessary to reduce the caliber of the instrument to increase the flow velocity in the pipe and obtain satisfactory measurement results.
3. Selection of screw connection vortex flowmeter transmitter
In the saturated steam measurement, the VA piezoelectric vortex flow transmitter is used. Because the vortex flowmeter has a wide range. Therefore, in practical applications, it is generally considered that the flow rate of the saturated steam should not be lower than that of the vortex flowmeter. Lower limit. That is to say, the fluid velocity must not be lower than 5m/s.
Vortex flow transmitters of different calibers can be selected according to the amount of steam used. The caliber of the transmitter cannot be selected by the caliber of the existing process pipeline.
4. Selection of pressure compensation pressure transmitter
Because the saturated steam pipeline is long, the pressure fluctuates greatly. Pressure compensation must be used. Considering the corresponding relationship between pressure, temperature and density, only pressure compensation can be used in the measurement. Because the pipeline saturated steam pressure is in the range of 0.3-0.7MPa. Pressure The range of the transmitter can be 1MPa.
5. Selection of display instrument for screw connection vortex flowmeter
Intelligent flow indicator, with functions of voltage stabilization compensation, instantaneous flow display and cumulative flow accumulation.
Further reading: Nitrogen(N2) Flow Meter Types
Digital Nitrogen Gas Flow Meters
Currently, on the market, there are four main instruments that can measure nitrogen flow:
Thermal gas mass flow meter. For direct mass measurement of industrial gases, compressed air, and aqueous fluids. Including nitrogen.
Coriolis mass flowmeter. Directly measure the mass flow of gas and liquid. It can be used to measure nitrogen, which is more expensive.
Vortex flowmeter. Directly measure the volume flow of gas and liquid. It is necessary to add temperature and pressure measuring sensors and totalizers to calculate mass flow.
Orifice flowmeter. The orifice is used as a throttling element. With a differential pressure transmitter, the volume flow can be calculated. To convert into mass flow, temperature and pressure sensors and totalizers must also be added.
Sino-Instt offer over 10 Threaded Vortex Nitrogen Gas Flow Meters, with Best Price.
A wide variety of Threaded Vortex Nitrogen Gas Flow Meters options are available to you, such as free samples, paid samples. All of our flow meters can wokrk with the AMS NEW TREX.
Sino-Inst is Threaded Vortex Nitrogen Gas Flow Meter suppliers, located in China. Vortex shedding flow meter products are most popular in North America, Mid East, and Eastern Europe.
Request a Quote