What Is Difference Between Rotameter and Flow Meter?
What is a rotameter flow meter?
Rotometer, also known as Float Type Flowmeter, is widely used in industry. It can measure the flow of liquid, gas and steam, and it is suitable to measure the flow of small and medium pipe diameters (DN4 ~ 250).
The rotor flowmeter measures the fluid flow according to the throttling principle. But it changes the flow area of the fluid to keep the differential pressure above and below the rotor constant. Therefore, it is also called a variable flow area constant differential pressure flowmeter, also known as a float flowmeter.
Transparent conical tube rotameter The most widely used material for transparent conical tube is glass. When the instrument without guide structure measures gas, the glass tube is easily broken. Transparent engineering plastics such as polystyrene and polycarbonate are also used. , plexiglass, etc., with the advantage of not easily broken.
Compared with the transparent conical tube rotameter, the metal tube conical tube rotameter can be used for higher medium temperature and pressure. And there is no potential danger of the glass tube rotameter cone being broken.
(1) Local indication type rotameter Some transparent tube rotameters are mainly based on local indication, and are equipped with proximity switches to output the upper and lower limit alarm signals of the flow. Some local indicating metal tube rotameters have the same shape as the remote signal output. Only the rotor displacement is transmitted through magnetic coupling. After being processed by a linearization mechanism such as a connecting rod cam, it is indicated on the spot.
(2) Remote signal output type rotameter The conversion part of the remote signal output type instrument converts the rotor displacement into current or air pressure analog signal output. They become electric remote rotameter and gas remote rotameter respectively.
It is divided into three types: liquid, gas and steam.
In fact, the same instrument of most rotameters can be used for liquid or gas, and the structure is universal.
It is just that the industry standard of my country’s rotor flowmeter (such as JB/T6844-93) stipulates that the upper limit of flow Qmax must meet the requirements of (1, 1.6, 2.5, 4 or 6) × 10nL/h (n is a positive or negative integer or zero). When the instrument designed for liquid (represented by water) is used for gas (represented by air), it does not meet the above requirements. Only a float and a conical tube can be designed for gas. It is divided into two series of liquid and gas.
In some manufacturers, the flow ranges of liquid and gas are paralleled in the same instrument. Of course, the flow values cannot be all rounded values. However, there are still differences in the design of liquid and gas. For example, the gas instrument rotor is designed to be lighter and prevent The structure of the damping member for the oscillation and beating of the float varies.
Only specially designed metal tube rotameters can be used to measure steam or additional components can be added to standard instruments. For example, liquid damping parts with heat sinks are added to reduce float jumps. Heat sinks are separated from the connection with the indication conversion part.
1) Full flow type That is, all the fluid to be measured flows through the meter of the rotameter.
2) Shunt type Compared with the full flow type, only part of the measured fluid flows through the flow detection part such as the rotor. The split-flow type rotameter is composed of a standard orifice plate (or an equal-velocity tube) on the main pipeline and a small-diameter rotameter. It is suitable for large flow with a pipe diameter greater than 50mm and places where local indication is required, and the price is low. The split flow rotor flowmeter is divided into two types: separate type and integrated type.
What Is Difference Between Rotameter and Flow Meter?
Q: What is the difference between an orifice flowmeter and a rotameter that measure the flow of media?
A: Simply put, the rotor flowmeter uses the gravity of the rotor to measure the flow, and the orifice plate uses the pressure difference before and after the fluid to measure the flow.
Rotameters measure fluid flow according to the throttling principle. But it changes the flow area of the fluid to keep the differential pressure above and below the rotor constant. Therefore, it is also called a variable flow area constant differential pressure flowmeter, also known as a float flowmeter.
Orifice flowmeter is a high range ratio differential pressure flow device composed of standard orifice plate and multi-parameter differential pressure transmitter (or differential pressure transmitter, temperature transmitter and pressure transmitter). It can measure the flow of gas, steam, liquid and natural gas.
Rotameter is a kind of flowmeter commonly used in industry and laboratories. It has the characteristics of simple structure, intuitive, small pressure loss, convenient maintenance and so on. The rotor flowmeter is suitable for measuring small flow through the pipe diameter D<150mm, and it can also measure the flow of corrosive medium.
The throttling device of the orifice flowmeter has a simple structure, is firm, and has stable and reliable performance. It has a long service life and a low price, and is a commonly used flow measuring instrument in the industry. The entire processing process adopts international standards and undergoes strict calibration and testing.
The working principle of the rotameter is to calculate the flow rate of the medium through the annular gap area between the float and the conical pipe. The turbine flowmeter calculates the flow rate of the fluid by proportional to the rotational angular velocity of the turbine and the fluid velocity. They are in There are also big differences in other respects.
Rotameter and turbine flowmeter are modern and new precision industrial measuring instruments. Although they are both used to measure the flow of media, they are different in working principle, manufacturing process, appearance structure, and performance. However, they all have Small size, simple structure, wide measurement range, long service life and other advantages. The only disadvantage of them is that they are only suitable for the measurement of fluid medium flow in small diameter pipes.
The difference between mass flowmeter and rotameter:
In terms of accuracy, the mass flowmeter can be controlled to 0.1ml/min,
The mass flow meter controls the pressure, and different pressure-resistant mass flow meters are selected according to different pressures.
Generally, the pressure resistance of the rotor flowmeter should be less than 10atm, while the mass flowmeter can reach 10MPa, and most of them are resistant to 10MPa.
The difference between a mass flowmeter and a rotameter is obvious. The mass flowmeter has high accuracy and may have a large pressure loss. Therefore, as long as the furnace ventilation is controlled, the rotameter is sufficient.
Yes, Rotameter is a commonly used flowmeter. It is suitable for liquid and gas volumetric flow measurement and control.
In a vertical conical tube that expands from bottom to top, the gravity of the circular cross-section float is supported by hydrodynamic forces. The float can rise and fall freely within the conical tube. It moves up and down under the action of flow rate and buoyancy. After balancing with the weight of the float, it is transmitted to the dial to indicate the flow through magnetic coupling.
Generally divided into glass and metal rotameters. Metal rotameters are the most commonly used in the industry. Glass materials are usually used for corrosive media with small diameters. Due to the fragility of glass materials, rotameters made of precious metals such as titanium are also used for key control points. .
No. Rotameters are not mass flowmeters.
Mass flowmeter and rotor flowmeter are both types of flowmeters. Mass flowmeter generally refers to a flowmeter that can directly measure mass flow readings.
The rotameter uses a float in a vertical conical tube to rise and fall with the flow change. A volumetric flow meter that changes the area of the circulation annulus formed between them to measure.
These two flowmeters are completely different, and users should not confuse them when using them.
Rotameters are devices that can measure the flow of liquids and gases.
Typically, a rotameter is a tube made of plastic, glass or metal combined with a float. It responds linearly to the fluid flow in this tube.
What is mechanical oil flow meter? Mechanical oil flow meter refers to a flowmeter that can detect the volume flow of oil inline. Most importantly, this type of flowmeter is a mechanical measurement. It is common to use an oval Read More
What is Clamp on Water Flow Meter? Clamp on Water Flow Meter is also known as the clamp-on ultrasonic flowmeter. Using an ultrasonic flowmeter to measure water flow rate is a great choice. Clamp on Water Flow Meter measures liquid Read More
Upstream and Downstream Flow Straight Pipe Requirements refer to the length of straight pipe before and after the installation location of the flowmeter. In order to ensure that the fluid flow state is uniform in the pipeline, the flow field Read More
Measuring Steam Flow Measuring Steam Flow is a difficult task. But again, it’s a very important thing. Steam flow measurement is difficult, the main reason is determined by the characteristics of steam. Steam is a relatively special medium. With the Read More
What is Digital Flow Meter for Argon Gas? Digital flow meter for argon gas refers to a flow meter that can be used for detection, display, transmission, and signal output. Digital argon flowmeter and argon regulator are different. Argon regulator Read More
A kerosene flow meter is a flow meter that can be used to measure the flow of kerosene. Kerosene is a common oil in industrial production. It is an organic solvent and has no electrical conductivity. Similar to diesel gasoline. Read More
Water flow measurement is common in both industry and life. You may often hear about the use of electromagnetic flowmeters to measure wastewater. Water flow velocity meter used at home. The clamp-on ultrasonic flowmeter measures large water pipes. The establishment Read More
Mechanical water flow meters refer to a type of flow meter suitable for clean water processes or applications. The working principle of a mechanical water flow meter is to measure the speed of water flowing through a pipe. This causes Read More
A flow totalizer meter is sometimes referred to as a flow meter or flow totalizer. flow totalizer meter displays the flow rate from either an analog or pulse output flow sensor as well as the accumulated total and grand total Read More
Ultrasonic flow detectors are an important part of ultrasonic flow meters. Ultrasonic flow detectors are also called ultrasonic transducers. Ultrasonic flow detectors measure the volumetric flow of liquid within closed conduits. Ultrasonic flow meters use transit-time differences for measuring flow Read More
A Venturi tube is a device used to measure the speed of a fluid. It can also be said to be a device used to measure the pressure difference of a fluid. Venturi tube is also often called Venturi meter, Read More
What is 4-20ma Flow Meter? 4-20mA Flow Meters are those flow meters with 4-20mA current signal output function. The signal current of general instruments is 4-20mA. This means that the minimum current is 4mA and the maximum current is 20mA. Read More
Molasses flow meters are a type of flow meter used for molasses flow measurement. Molasses are common high viscosity liquids. So Molasses flow meters are flow meters that can be used to measure various high viscosity liquids. Can common electromagnetic Read More
What is a low flow meter? Low Flow Meters refer to those flow meters that can be used to measure micro-flow-low-flow rate media. Including metal rotor type, gear type, electromagnetic type microflow meter, microturbine type, and thermal type small mass Read More
Differential pressure flow meters can measure all single-phase fluids. It can measure liquid, gas, and steam. Such as gas-solid, gas-liquid, liquid-solid, etc. can also be applied. The calculation work in the use and maintenance of the differential pressure flow meter Read More
What is a float flow meter? The float flow meter is also called a rotameter. The float flow meter is a volumetric flow meter in which a float rises and falls in a vertical tapered tube as the flow changes, Read More
A gear flow meter is a common positive displacement flow meter. The gear flow sensor has built-in double gears to operate. The volume of the medium is calculated by the volume of the gear. The flow rate of the small Read More
What is a Grease Flow Meter? Grease Flow meters are meters dedicated to measuring the volume or mass flow of industrial grease fluids. The most common Grease Flow meters are elliptical gear flow meters, also called positive displacement flowmeters. There Read More
What is meant by Modbus protocol? Modbus Protocol is a serial communication protocol, which was published by Modicon company (now Schneider Electric) in 1979 for the use of programmable logic controller (PLC) communication. Modbus has become the industry standard (De Read More
Sino-Inst offers over 30 Rotameter flow meter products. About 50% of these are for liquids. 40% are for gas. At best price.
You can refer to the product price of the following Rotameter flow meter price:
Metal rotor flowmeter Pipe size 1 inch FOB Price: USD 381.50/pc~USD699.00/pc;
A wide variety of Rotameter flow meter options are available to you, such as free samples, paid samples.
Sino-Inst is a globally recognized supplier and manufacturer of Rotameter flow meters, located in China.
The top supplying country is China (Mainland), which supply 100% of the Rotameter flow meter respectively.
Sino-Inst sells through a mature distribution network that reaches all 50 states and 30 countries worldwide. Rotameter flow meter products are most popular in Domestic Market, Southeast Asia, and Mid East.
You can ensure product safety by selecting from certified suppliers, with ISO9001, ISO14001 certification.