What is 4-20ma Flow Meter?
4-20mA Flow Meters are those flow meters with 4-20mA current signal output function. The signal current of general instruments is 4-20mA. This means that the minimum current is 4mA and the maximum current is 20mA. Our commonly used electromagnetic flowmeters, turbine flowmeters, mass flowmeters, gas flowmeters, etc. all have the function of 4-20mA signal output.
What is a 4-20mA Output？
The signal current of general instruments is 4-20mA, which means that the minimum current is 4mA and the maximum current is 20mA.
The industry generally needs to measure various non-electrical physical quantities. For example, temperature, pressure, speed, angle, etc. They all need to be converted into analog electrical signals before they can be transmitted to the control room or display equipment hundreds of meters away. This device that converts physical quantities into electrical signals is called a transmitter. The most widely used in the industry is to use 4-20mA current to transmit analog quantities.
In the industrial field, to complete signal conditioning and long-term transmission, the following problems will occur:
- Since the transmitted signal is a voltage signal, the transmission cable will be interfered by noise;
- The distributed resistance of the transmission cable will produce a voltage drop;
- How to provide the working voltage of the instrumentation amplifier in the field is also a problem.
In order to solve the above problems and avoid the influence of related noise, we will use current to transmit signals. Because the current is not sensitive to noise.
The 4-20mA current loop uses 4mA to represent the zero signal. Use 20mA to represent the full scale of the signal. The signals lower than 4mA and higher than 20mA are used for various fault alarms.
Why Choose 4-20mA Instead of 0-20mA for the Flowmeter?
Many people will certainly ask: Why choose 4-20mA instead of 0-20mA?
Very simple, if 0 is the smallest, then the open circuit fault will not be detected!
So, why is it 4mA?
During normal operation, the current signal will not be lower than 4mA.
When the transmission cable is broken due to a fault, the loop current drops to zero. 2mA is often used as the disconnection alarm value.
There are two reasons. One reason is to avoid interference. Another reason is that a two-wire system is used at 4-20mA. That is, the two wires are the signal wires. It is also a power cord. And 4mA is to provide the static working current of the circuit to the sensor.
Extended Reading: Mechanical flow meter for diesel
How Does This 4-20mA Control Loop Work?
4-20mA constitutes the basic elements:
- 24V power supply
- The transmitter controls the 4-20mA signal to change in proportion to the process variable
- The indicator converts the 4-20mA signal into the corresponding process variable
- Indicator or controller I/O input resistance 250Ω shunt generates 1-5V input signal (Ohm’s law: voltage = current * resistance, 4-20 mA X 250 ohms = 1-5V)
- They convert the temperature signal of the thermocouple or thermal resistance sensor into a 4-20mA signal and then output;
- The controller then translates 4-20mA into a specific temperature value;
- Based on this temperature value. The control loop realizes the control of the process terminal control element.
Similarly, the pressure transmitter is in the control loop. Usually used to measure the pressure value of the process medium:
1) The sensor senses pressure. The transmitter converts the signal into 4-20mA signal;
2) The controller then reversely translates the 4-20mA signal into a pressure value;
3) The controller is based on the pressure value. Send instructions to the valve. Control valve opening to realize safety valve control. Ensure that the container does not generate dangerous pressure.
Extended reading: Orifice Plate Flow Meter
Water Flow Meter 4-20ma Output
A Digital Water flow meter is a flow meter that can display the water flow rate, and have the output for the flow rate. Like 4-20mA output, HART, RS 485, and so on. Digital Water Flow Meter is also called Electronic Water Flow Meter. Digital Water Flow Meter is widely used in industrial water and household water measurement. Most of the Sino-Inst flow meters are digital water flow meters. Like the electromagnetic flowmeter, Portable ultrasonic flow meter, Turbine flow meter…..
No matter which WaterFlow Meters you choose. The signal output by WaterFlow Meters, such as 4-20mA. The flow signal can be connected to a paperless recorder. Carry out multi-channel flow signal monitoring.
Read More About: Paperless Recorder Working Principle
Flow Meter with Analog Output
The analog flow meter is a flowmeter with the analog output signal. Analog output is a common function of flow meters. The analog signal is mainly used to output the flow signal to the next control system.
The analog signal refers to the continuous signal of the information parameter in a given range. The characteristic quantity of the information of the analog signal can be transformed into an arbitrary numerical signal in an instant. In the process of transmitting analog signals, the information signals need to be converted into electric wave signals, which are then transmitted through wired or wireless methods.
Analog signal: poor confidentiality, weak anti-interference ability.
Digital signal: strong anti-interference ability and good communication confidentiality.
Read more about Analog Flow Meters.
Turbine Digital Flow Meter
Turbine flow meters are velocity flow meters, also called impeller flow meters. Can be used to measure the instantaneous flow and cumulative flow of liquids and gases.
Turbine flowmeters are widely used in flow measurement: petroleum, organic liquids, inorganic liquids, liquefied gas, natural gas, coal gas, and cryogenic fluids. Turbine flowmeter signals can be divided into pulse signals or current signals (4-20mA). It is suitable for use with secondary display, PLC, DCS, and other computer control systems.
Turbine Flow Meter reference price: USD 200-1700/pc.
4-20ma Air Flow Meter
A digital air flow meter refers to a flow meter that can accurately measure air or compressed air inline. Industrial Digital air flow meters can be divided into mass flow meters or volume flow meters.
Read more about Digital air flow meters.
Ultrasonic Flow Detectors
Flow Rate and Pressure
What is a Venturi Tube?
Molasses Flow Meters-High Viscosity Liquids Solution
Solvent flow meters
Grease Flow Meters
3 inch (3″) Water Meter
2 inch Water Flow Meter
Industrial CO2 flow meters
Looking for 4-20ma flow meters? The flow meter is a fast, accurate, and reliable permanent linear motion. In Sino-Inst, you can also find other good deals on measurement and analysis instruments and tools! Keep an eye out for promotions and deals, so you get a big saving of 4 20ma flow meters. You can shop for 4 20ma flow meters at low prices.
Extended reading: How to convert a 4-20mA to 0-10V /1-5V signal?
There is a wide variety of 4-20ma flow meter items you can buy, such as turbine water flow meters, ultrasonic flowmeters. Common materials used for 4 20ma flow meters are stainless steel and carbon steel.
Sino-Inst offers over 50 4-20ma flow meters for flow measurement. About 50% of these are differential pressure flow meters, 40% are the liquid flow sensor, and 20% are Ultrasonic Level Transmitter and mass flow meters.
A wide variety of flow meters options are available to you, such as free samples, paid samples.
Sino-Inst is a globally recognized supplier and manufacturer of flow measurement instrumentation, located in China.
Request a Quote